Introduction The pipeline embolization device-(PED) has gained popularity in the treatment of cerebral aneurysms in selected patients, with studies evaluating its efficacy and safety showing promising results. Nonetheless, difficulty persists with deployment in tortuous vasculature, with less than ideal stent positioning, stent stretching, and poor stent/arterial wall apposition encountered in such instances. We report our experience with balloon angioplasty in reducing stent stretching, improving stent/arterial wall apposition, and aiding in stent repositioning.
Methods A retrospective medical record, angiographic and video review of all pipeline cases done at the University of Illinois was performed, and cases where balloon angioplasty was utilized were selected. Vessel tortuosity was classified according to the published method by Lin et al.1
Results A total of 6 patients were identified. All had pipeline stent deployment in the cavernous/paraclinoid segment of the ICA. We found (3) type-IV, (2) type-III, and 1 type-II ICA as classified by Lin et al.1 Indications for balloon angioplasty were incomplete stent apposition in 5 cases, and poor stent position in 1 case. A balloon was successfully used in all 6 cases, improving stent/arterial wall apposition. In all cases, a Transform balloon was used over exchange wire to maintain endoluminal access. Additionally, proximal stent stretching was observed in two of the six cases, and successfully reduced with the balloon, resulting in improved positioning and apposition of the stent. Four patients have had follow-up angiogram at 6 months, none of which showed in-stent stenosis.
Conclusions Tortuous ICA vasculature can increase the risk of stent stretching, incomplete stent/arterial wall apposition, and poor stent positioning. Balloon angioplasty is a safe and effective method to circumvent these challenges, and should be considered as a viable option in these cases.
Lin LM, Colby GP, Jiang B, et al. Classification of cavernous-ICA tortuosity: a predictor of procedural-complexity in Pipeline-embolization. J Neurointerv Surg. 2014
Disclosures Z. Hage: None. A. Alaraj: None. V. Aletich: None.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.