The pipeline embolization device (PED) has become a routine firstline option for the treatment of an increasing population of intracranial aneurysms at many neurovascular centers. Intraprocedural complications during PED deployment, or complications associated with migration of the device, are rarely reported problems. Significant mismatch in luminal diameter between the inflow vessel and the outflow vessel or excessive dragging–stretching of the PED have been related to ‘watermelon seed’ or ‘accordion’ effects, respectively, resulting in stent migration. Here we present a novel balloon technique that was successfully used to realign an in situ flow diverting stent that had prolapsed into a large aneurysm. This represents a useful salvage technique and should be considered when encountering this potential complication.
- Flow Diverter
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.