Background Although aspiration and stent retriever thrombectomy perform similarly in proximal occlusions, no comparative series are available in distal occlusions. We aimed to compare the 3 mm Trevo Retriever against the 3MAX thromboaspiration catheter in distal arterial occlusions.
Methods A single-center retrospective review of a prospectively maintained databank for patients treated with the 3 mm Trevo stent retriever or 3MAX thromboaspiration as the upfront approach for distal occlusions (middle cerebral artery mid/distal M2/M3, anterior cerebral artery A1/A2/A3 or posterior cerebral artery P1/P2) from January 2014 to July 2018 was performed. The primary outcome was the rate of distal occlusion first-pass reperfusion (modified Thrombolysis in Cerebral Infarction (mTICI) 2b–3).
Results Of 1100 patients treated within the study period, 137 patients/144 different arteries were treated with the 3 mm Trevo (n=92) or 3MAX device (n=52). The groups had comparable demographics and baseline characteristics. There was a higher rate of first-pass mTICI 2b–3 reperfusion (62% vs 44%; p=0.03), a trend towards a higher rate of final mTICI 2b–3 reperfusion (84% vs 69%; p=0.05), and lower use of adjuvant therapy (15% vs 31%; p=0.03) with the 3 mm Trevo compared with the 3MAX. The median number of passes (p=0.46), frequency of arterial spasm (p=1.00), rates of parenchymal hematomas (p=0.22)/subarachnoid hemorrhage (p=0.37) in the territory of the approached vessel were similar across the two groups. The 90-day rate of good outcomes (45% vs 46% in the 3 mm Trevo and 3MAX groups, respectively; p=0.84) was comparable. Multivariable regression identified baseline NIH Stroke Scale (NIHSS) score (OR 0.9; 95% CI 0.8 to 0.97; p<0.01) and use of 3 mm Trevo (OR 2.2; 95% CI 1.1 to 4.6; p=0.02) independently associated with first-pass mTICI 2b–3 reperfusion.
Conclusions In the setting of distal arterial occlusions, the 3 mm Trevo may lead to higher rates of first-pass reperfusion than direct 3MAX thromboaspiration. Lower NIHSS was found to be associated with improved reperfusion rates as observed in more proximal lesions. Further studies are warranted.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
Contributors DCH: Study conception, design of the work, acquisition of data, statistical analysis, interpretation of data, drafting of the manuscript. BE, ARA, JAG, GMR, MRF: Data acquisition, critical revision of manuscript. RN: Design of the work, acquisition of data, interpretation of data, critical revision of manuscript.
Funding The authors have not declared a specific grant for this research from any funding agency in the public, commercial or not-for-profit sectors.
Competing interests DCH: Consultant for Stryker and Vesalio. RN: Principal Investigator, Stryker Neurovascular (DAWN trial (no compensation), Trevo‑2 trial), Cerenovus/Neuravi (ENDOLOW trial, no compensation); consultant to Stryker Neurovascular; steering committee member, Stryker Neurovascular (no compensation), Medtronic (SWIFT trial, SWIFT Prime trial (no compensation)), Cerenovus/Neuravi (ARISE‑2 trial, no compensation); angiographic core lab, Medtronic (STAR trial); executive committee member, Penumbra (no compensation); physician advisory board, Cerenovus/Neuravi, Phenox, Anaconda, Genentech, Biogen, Prolong Pharmaceuticals, Allm (no compensation), Viz-AI; stock options, Viz-AI.
Ethics approval Emory University Institutions Review Board.
Provenance and peer review Not commissioned; externally peer reviewed.
Data sharing statement The unpublished data from this dataset is held by Grady Memorial Hospital/Emory University and DCH/RN. Requests for data sharing would be required to be discussed with them directly.
Patient consent for publication Not required.